The Instruction of LS126C Ultraviolet Irradiance Meter
The LS126C Ultraviolet Irradiance meter is specially used to detect the irradiation intensity of 254nm germicidal lamp. The instrument is equipped with a standard 1 meter test hook and can be used without protective equipment. Then what is the operation method of LS126C Ultraviolet Radiation Meter?
1. Assemble the hooks on the instrument, as shown below:
2. Press the “POWER” button on the left side of the instrument, it will display a set of data, and in the lower left-hand corner of the screen display No.1. This is the latest test data of the
3. Press the “HOLD” button in the middle of the instrument to unlock the test on the instrument, the “No.1” mark disappears and the max value and RT value on the screen return to zero.
4. Place the ultraviolet radiation meter on a fixed position of the hook, and the make the hook hang under the ultraviolet germicidal lamp.
5. Turn off other fluorescent lights in the room and close the curtains, power on the ultraviolet germicidal lamp which need to be tested. And all the people can leave the room.
6. After about 20 minutes, turn off the ultraviolet light and get the instrument. At this time the instrument has been already shut down automatically, press the “POWER” button to boot, the screen displays the “No.1” set of data, which is the test results.
According to the sterilization Lamp detection standard, LS126C Ultraviolet Irradiance Meter is equipped with a 1m test hooks, and the instrument will shutdown automatically after 10 minutes, it can test without people and any protective equipment.But we also need to ensure that only the ultraviolet light is on during the measurement process, no other external light,on in this way, the test data will be more accurate.
- Attentions in Type Selection of254nm UVC Light Meter2018-12-10
- Advantages of LS110H and LS110A Split Transmission Meter2018-11-27
- The PC Software Function of Marble Gloss Meter2018-11-23
- Test Glassine Transmittance by Diffuse Transmission Material Transmittance Meter2018-11-15
- LS117 Optically Diffuse Materials Transmittance Meter2018-11-06
- The Calibration and Maintenance of UV Light Meter and UV Energy Meter2018-11-02