Attentions in Type Selection of 254nm UVC Light Meter
Linshang LS126C 254nm UVC light meter has a 10-minute automatic shutdown function, equippedwith a 1-meter ruler hook make it simple to detecting UV germicidal lamps.
Linshang LS126C is an UV irradiation meter specializing in the detection of 253.7nm UVCgermicidal lamps, LS126C has the following advantages when compared to othertypes of UV germicidal lamps irradiation meters.
1. 254nm UVC irradiation meter is equipped with a 1-meter hook, whentesting it can be put on the hook to make the measurement more convenient.
2. The maximumand real-time values during the detection process can be recorded at the sametime, after the test we can determine whether the germicidal lamp is qualifiedaccording to the maximum value.
3. Theinstrument has a 10-minute automatic shutdown function and can display the lastmeasurement results after the next boot. The tester can leave the scene duringthe test to avoid being harmed by ultraviolet rays.
Attentions inType Selection of 254nm UVC Light Meter
1. The firststep is to confirm the type of UV light source being measured, and differenttypes of light sources need to be used with different UV irradiation meters. Ultravioletrays are mainly divided into UVA, UVB and UVC.
2. If it isconfirmed that it is a UVC light source, it is necessary to reconfirm whetherthe wavelength of the UV light source is within the spectral response range ofthe 254nm UVC irradiation meter, and the spectral response range of the LS126CUVC light meter is 200nm-280nm, if the wavelength of UV light source is not inthis range, The data measured by the instrument will be inaccurate.
3. Finally it isnecessary to confirm the intensity range of ultraviolet light source, differentUV irradiation meter. The range is not the same. The measurement range of theLS126C UVC UV light meter is 0-8000uw/cm2, and if this range is exceeded, the LS125 UV irradiation meter and the P254 UV probe can be considered.
- The Difference between Upgraded and Old LS125 UV Light Meter2019-02-16
- The Principle of Linshang UV Energy Meter2019-01-28
- The Maintenance of Linshang Gloss Meter 2018-12-20
- Advantages of LS110H and LS110A Split Transmission Meter2018-11-27
- The PC Software Function of Marble Gloss Meter2018-11-23
- Test Glassine Transmittance by Diffuse Transmission Material Transmittance Meter2018-11-15